Relative humidity: not more than 30% 70%. Transport & storage temperature: -20. 93% Atmospheric pressure range: 500hPa 1060hPa. Before the operation, please confirm the following information. Ensure that the water level accord with the description (see 2-5-3).
Check all the connection of the system. Ensure that all personnel in the area are wearing protective glasses. Make sure the emergency shut-off knob is loosed. After these steps, turns the key clockwise, the waterway and screen begin to work, the system get start.
In normal situation, you can hear the cycling of the inner waterway. Skin test is essential before the optional treatment.The details are as following. If do treatment for all face, the test patch should be selected under the ears where not influencing the facial beauty. If for chest, neck or other skin, it should be selected the site where not influencing the beauty.
Clean the test patch and apply a layer of coupling gel at the test patch. Take out of the treatment handle and aim the light guide at the test patch.
Trigger 2-3 shots 1/3 area of the spot size overlapping the treatment site is permitted. , and wait for 2-3 minutes to examine the test patch.
If lightly swelling and/ or reddening at the test patch appears several minutes later, continue the treatment with the same settings of energy level and pulse type. If no visible changes at the patch appear, continue the test with 1-2j/cm² higher energy level. If the skin show adverse effect or excessive reddening and/or swelling at the test patch just after shot, continue the test with lower energy level. Must protect patients eyes when the system is in use.Start the treatment from the ear-root, and spread to all around. Place the treatment handle to the skin about 1-3mm, but not apply pressure. Leave no blank site and overlapping 1/3 of the spot size is allowed. Do not shoot the light at upper eyelid. Change the direction of the treatment head so that it can treat more evenly and roundly.
Lengthen D1, D2 for darker skin. Shorten D1, D2 for light skin.
Make sure the emergency shut-off knob loosed, turns the key clockwise, the waterway and screen begin to work, the system get start. Is used for hair removal.Can cure pigmented spots on the superficial of skin. The operator can select suitable treatment mode according to clients requirements. Here take SHRmode as example. Both of these operations are same. The system will change into the following working interface. Hand-touch Next , the system will into the following working interface. The system have three programs, each of them has respective parameters. User can touch Program 1, 2, 3 and , + to select suitable energy and parameters. Cancel the number recorded in this operation. Off the lamp in the treatment head. This is the skin type classification according to customer's skin color to choose, one is white, two kinds of light colored race. The yellow race, class four is brown, the five category is the black race. The highest energy 50J, starting 5J, according to the customer the degree of tolerance of 2J step, "+" or "-". The highest frequency is 10HZ, recommend the use of 3-5HZ, it can prolong the service life of head. Function respectively, temperature display, water. Here take E-light mode as example. Next , the system will into the following working interface.
Enter the E light option, there will be six types of treatment mode, according to the different needs of the patients, the choice of different treatment. Can cure the corresponding selection differential. To choose the treatment project according to the demand of different treatment of. The classification of a class of skin, white, two kinds of the yellow race, class three is brown, the four category is. The highest energy RF IPL is 50J, the highest energy is 50J, the general IPL energy start is 10J, according to the guests to feel "+", "-" general step is 2J, RF energy in general is adjusted to 15J can, do not need to adjust!Touch Standby on the screen , then turn the key switch to Off. Clean the treatment light guide with alcoholic cotton ball. Construction of the system shows above: color LCD screen, emergency shut-off knob, key switch, main console, treatment handle plug, treatment handle etc. Please take necessary protections for the operation of the machine and handle piece, and notice that following situations can not be warranted. Operation the system without water. Power up the system without connecting the handle piece.
Regular drops of the handle piece. Clean the system exterior at least once a week. Wipe all surfaces with soft and damp cotton ball.Not spill liquids in the machine. The light guide should keep clean at all times and should be cleaned after each patient. To clean the light guide, first dry it with a lint-free cloth. Then clean it with a cloth moistened with water-free alcohol and let it dry thoroughly. If you are satisfied with the item, pls.
If you are not satisfied/received, pls. We will try our best to solve the problems for you. If you have this need, please contact me as soon as possible!The item "2in1 permanent hair removal SR+HR OPT machine/e-light IPL RF laser multifunction" is in sale since Tuesday, May 05, 2015. This item is in the category "Health & Beauty\Shaving & Hair Removal\Laser Hair Removal & IPL". The seller is "xiranzheng" and is located in , . This item can be shipped worldwide.